100mA pet X-ray machine/bedside machine
The main technical parameters:
1. Power conditions and operation mode
Power supply voltage: AC 220V±22V;
Power frequency: 50Hz±0.5Hz;
Power capacity: ≥8KVA;
The maximum allowable internal resistance of the power supply: 1Ω
Operation mode: intermittent loading continuous operation
2. The maximum rated capacity is shown in Table 1
Table 1. Maximum Rated Capacity
Tube current (mA) Tube voltage (kV) Time (s)
15 90 6.3
30 90 6.3
60 90 4.0
100 80 3.2
3. Photography conditions: Tube voltage: 50-90kV
Tube current: 15, 30, 60, 100mA in 4 gears;
Time: 0.08s-6.3s, a total of 19 gears, selected according to the R10' coefficient.
4. Maximum output power:
( 80kV 100mA 0.1S ) 5.92KVA.
5. Nominal electric power:
( 90kV 60mA 0.1S ) 4.00KVA.
6. Input power: 5.92KVA.
7. Mechanical properties:
When the X-ray generator window is downward, the distance between the focus and the film is 1000mm;
The rotation angle of the X-ray generator around the vertical axis is ±90º;